Biological safety cabinets (BSCs) are used at the University as a primary means of
containment for working safely with infectious microorganisms. A properly certified
and operational biological safety cabinet is an effective engineering control which
must be used in concert with appropriate work practices, procedures and other administrative
controls to reduce the risk of exposure to infectious microorganisms. The Biosafety
Office can provide guidance on the selection, installation and use of biological safety
Biological safety cabinets must be tested and certified at least annually to ensure
continued proper operation. All Class II BSCs will be tested and certified in accordance
with specifications in NSF Standard 49 and the manufacturer’s specifications. The
NSF 49 establishes minimum requirements for materials, design, construction, and performance
of biosafety cabinets that are designed to provide personnel, product, and environmental
protection. The operational integrity must be validated by certification before a
newly installed BSC is used and after a BSC has been repaired or relocated. After
a BSC has been certified, a label will be prominently affixed to the front of the
BSC, displaying the date of certification and name of the certifier.
BSCs that have been used for work involving infectious materials must be decontaminated
before HEPA filters are changed or internal repair work is done. BSCs must be decontaminated
prior to decommissioning and salvaging. Before a BSC is relocated, a risk assessment
considering the agents manipulated within the BSC must be performed to determine the
appropriate method for decontamination.
The Biosafety Office manages the University’s contract for the annual certification of all biological safety cabinets. Technical Safety Services is the approved vendor for performing the annual certification of biological safety cabinets, including any required maintenance or repairs. You can contact the vendor directly at:
Technical Safety Services, LLC
BSC Moving, Lifespan and Decommissioning
It is a common practice to move permanently installed biosafety cabinets (BSCs) to other locations within a laboratory or to other laboratories. BSCs should not be moved without consultation with the Biosafety Office.
The current lifespan of a BSC is approximately 15 years. After 15 years, replacement parts may or may not be available due to electrical or mechanical changes at the factory or industrial part suppliers. BSCs have evolved through the years with many improvements in containment, ergonomics, serviceability, and energy efficiency. These issues should be considered when making decisions on BSC repair versus replacement.
No BSC should be sent to a landfill or a recycling facility as a BSC, it should be
disassembled per requirements contained in the NSF/ANSI 49 Standard – Annex E (Section
E.10). Labs must consult with the Biosafety Office prior to decommissioning a BSC.